Payment & Security
Woodpecker's Ai-Pex - Apex Locator with Pulp Tester
Can be connected to Endo Motor
Constant current source
Electric pulp testing via constant current rather than voltage tester.
When the clinical needs to determine the vitality of the root pulp, the electric pulp test can be a useful adjunct.
In the case of thermal pulp test, the cold and heat stimuli can only reach the coronal pulp.
Constant Current can assist with additional measurements
Magnetic adsorption base
• Unique base design.
• 360º rotatable.
• The viewing angle of the screen can be adjusted to elevate external glare or to improve visibility.
Multi-frequency DSP technology
Large colour screen
• 3.8" LCD touch screen with clear accurate colours.
• Interface uses four main colours to display depth.
• It is clinically recommended to make the measurement to the 0.0 position, then manually subtract 0.5~1mm to obtain the final length.
• Enables easier measurement.
• When the length indicator bar reaches a digital number below 1.5, the device emits a low-frequency series of beeps.
• The speed of the beeps increases when the tip is closer to the apical foramen.
• Once the tip reaches the apical stop, the device emits a continuous high-frequency beep.
• The sound level can be adjusted as needed.
Root Canal Length Measurement
• Auto calibration after startup.
• New digital circuit.
• U.S. central processing unit with DSP processing module.
• 0.1% high precision components.
• Information processing speed increased by 2.2x
Enable working length control during preparation when connected to an Endo Motor
Connecting the Ai-Pex to an Endo Motor that can be connected to an Apex Locator allows for real-time observation of the position of the file in the root canal during operation, and activates enabled automatic functions (e.g. Auto Reverse)
Accurate measurement of root canal length
• Lowered environment requirements for root canal length measurement
• Suitable for various complex root canals
• Unaffected by blood and residual pulp, whether the root canal is dry or wet, it can automatically and precisely determine the position of the apical foramen.
How to use the pulp vitality test function:
• Long press the function button to switch to PT mode.
• Press the speed button to adjust to the suitable working mode.
• Connect lip hook and probe.
• Coat the head of the probe with a layer of conductive agent (toothpaste, etc.)
• Place the probe at the 1/3 position of the tooth neck.
• Test the tooth and it's adjacent tooth separately and compare the values when the patient feels a tingling sensation.
• Determine the pulp vitality based on comparative numerical analysis.
Screen: 3.8-inch LCD display
Charging time: 3 hours
Use time: 5 hours 20 minutes
Battery Capacity: 2000mAh
Power consumption: ≤0.5W
Input ~100V-240V 50Hz/60Hz 0.4A
This Alex locator is not just ultra-modern to fit in with any new fashionable practices but I’ve found it reliable for all cases so far.
It’s compactness and portability are also major positives as I travel between practices for Endo.
Highly recommend this product. Having used many apex locators I can definitely say that this is one of the easiest and most reliable. The function as a pulp tester is very useless and wouldn’t hesitate to recommend this to anyone thinking about purchasing this.